Hi!

I need help to modify the usrp_fft.py file to show the spectrum of a

input

range (for example from 2.3GHz to 2.9GHz)

santi, such a wide frequency range is not possible due to limitations

in the bandwidth between the usrp and the host

currently you give usrp_fft a center frequency and a decimation rate,

and from that it calculates the start and end frequencies…

Dimitris S.

“If you think you’re too small to make a difference, try sleeping with

a mosquito!” - Amnesty International

Yes, but I think we can take the spectrum of every 8MHz band and put it

into

the computer so it can memorize and show it.

Sorry for my english…

2008/10/14 Dimitris S. [email protected]

Hi,

Santi O. [email protected] wrote:

Yes, but I think we can take the spectrum of every 8MHz band and put it into the

computer so it can memorize and show it.

See usrp_spectrum_sense.py, it implements this technique to scan the

entire spectrum available to each USRP daughter board.

Regards,

Firas

On Tue, Oct 14, 2008 at 11:33 AM, Santi O.

[email protected] wrote:

ok, but this program doesn’t show you anything… just uOuOuO…

Please reference:

```
http://gnuradio.org/trac/wiki/UsrpFAQ/Gen#OUuainoutput
```

Brian

Hi,

Santi O. wrote :

ok, so I have to modify this program to show in a Frame the results I

want, haven’t I?

Yes, OR, you can wait for me to write a detailed explanation (may be in

two

weeks) about “usrp_spectrum_sense.py”.

I think the understanding of this program is very important to every

gnuradio user (practical FFT implementation + FSM control in its cpp

code

and weird Python commands in its .py!!!)

Best regards,

Firas

–

View this message in context:

http://www.nabble.com/Wideband-Spectrum-Analyzer-tp19974701p19977090.html

Sent from the GnuRadio mailing list archive at Nabble.com.

ok, so I have to modify this program to show in a Frame the results I

want,

haven’t I?

2008/10/14 Brian P. [email protected]

Firas,

have to written a detailed explanation about “usrp_spectrum_sense.py” ?

I

am unable to find it. Please let us know the link for the same…

Shesh

Firas A. wrote:

–

View this message in context:

http://www.nabble.com/Wideband-Spectrum-Analyzer-tp19974701p24162627.html

Sent from the GnuRadio mailing list archive at Nabble.com.

shesh wrote:

Firas,

have to written a detailed explanation about “usrp_spectrum_sense.py” ? I

am unable to find it. Please let us know the link for the same…Shesh

## http://www.nabble.com/Some-usrp_spectrum_sense.py-code-Explanation-td21209623.html#a21209623

View this message in context:

http://www.nabble.com/Wideband-Spectrum-Analyzer-tp19974701p24163935.html

Sent from the GnuRadio mailing list archive at Nabble.com.

Hello

I wonder someone can help me out. While testing this code, the media

with the different size of fft shows different value. I would like to

know how I can get the same media value regardless of using different

fft size.

Thanks,

Kyungtae-- Santix wrote :

```
media=str(mean(m.data)) #
todo= p + " " + media + '\n' #
power.write(todo) #
```

–

This message was sent on behalf of [email protected] at

openSubscriber.com

http://www.opensubscriber.com/message/[email protected]/10657497.html

Hi everybody!

I have modified usrp_spectrum_sense.py to plot the results with gnuplot.

There are two files: widespectrum.py and plot.p

I would like everybody to test it and report me the errors and how can I

improve it.

I’ve used USRPv1 + Flex2400.

Thanks in advance!

Here it goes…

WIDESPECTRUM.PY:

#!/usr/bin/env python

# Copyright 2005,2007 Free Software Foundation, Inc.

# This file is part of GNU Radio

# GNU Radio is free software; you can redistribute it and/or modify

# it under the terms of the GNU General Public License as published by

# the Free Software Foundation; either version 3, or (at your option)

# any later version.

# GNU Radio is distributed in the hope that it will be useful,

# but WITHOUT ANY WARRANTY; without even the implied warranty of

# MERCHANTABILITY or FITNESS FOR A PARTICULAR PURPOSE. See the

# GNU General Public License for more details.

# You should have received a copy of the GNU General Public License

# along with GNU Radio; see the file COPYING. If not, write to

# the Free Software Foundation, Inc., 51 Franklin Street,

# Boston, MA 02110-1301, USA.

from gnuradio import gr, gru, eng_notation, optfir, window

from gnuradio import audio

from gnuradio import usrp

from gnuradio.eng_option import eng_option

from optparse import OptionParser

from usrpm import usrp_dbid

import sys

import math

import struct

import Gnuplot, Gnuplot.funcutils # Added to view the results

class tune(gr.feval_dd):

“”"

This class allows C++ code to callback into python.

“”"

def **init**(self, tb):

gr.feval_dd.**init**(self)

self.tb = tb

```
def eval(self, ignore):
"""
This method is called from gr.bin_statistics_f when it wants to
```

change

the center frequency. This method tunes the front end to the

new

center

frequency, and returns the new frequency as its result.

“”"

try:

# We use this try block so that if something goes wrong from

here

# down, at least we’ll have a prayer of knowing what went

wrong.

# Without this, you get a very mysterious:

#

# terminate called after throwing an instance of

‘Swig::DirectorMethodException’

# Aborted

#

# message on stderr. Not exactly helpful

```
new_freq = self.tb.set_next_freq()
return new_freq
except Exception, e:
print "tune: Exception: ", e
```

class parse_msg(object):

def **init**(self, msg):

self.center_freq = msg.arg1()

self.vlen = int(msg.arg2())

assert(msg.length() == self.vlen * gr.sizeof_float)

```
# FIXME consider using Numarray or NumPy vector
t = msg.to_string()
self.raw_data = t
self.data = struct.unpack('%df' % (self.vlen,), t)
```

class my_top_block(gr.top_block):

```
def __init__(self):
gr.top_block.__init__(self)
usage = "usage: %prog [options] min_freq max_freq"
# Example: ./widespectrum.py 2.23G 2.93G
# that is the maximun range of the USRP Flex2400 device.
parser = OptionParser(option_class=eng_option, usage=usage)
parser.add_option("-R", "--rx-subdev-spec", type="subdev",
```

default=(0,0),

help=“select USRP Rx side A or B (default=A)”)

parser.add_option(“-g”, “–gain”, type=“eng_float”,

default=None,

help=“set gain in dB (default is midpoint)”)

parser.add_option(“”, “–tune-delay”, type=“eng_float”,

default=1e-3, metavar=“SECS”,

help=“time to delay (in seconds) after

changing

frequency [default=%default]”)

parser.add_option(“”, “–dwell-delay”, type=“eng_float”,

default=10e-3, metavar=“SECS”,

help=“time to dwell (in seconds) at a given

frequncy [default=%default]”)

parser.add_option(“-F”, “–fft-size”, type=“int”, default=256,

help=“specify number of FFT bins

[default=%default]”)

parser.add_option(“-d”, “–decim”, type=“intx”, default=64,

help=“set decimation to DECIM

[default=%default]”)

parser.add_option(“”, “–real-time”, action=“store_true”,

default=False,

help=“Attempt to enable real-time scheduling”)

parser.add_option(“-B”, “–fusb-block-size”, type=“int”,

default=0,

help=“specify fast usb block size

[default=%default]”)

parser.add_option(“-N”, “–fusb-nblocks”, type=“int”, default=0,

help=“specify number of fast usb blocks

[default=%default]”)

```
(options, args) = parser.parse_args()
if len(args) != 2:
parser.print_help()
sys.exit(1)
self.min_freq = eng_notation.str_to_num(args[0])
self.max_freq = eng_notation.str_to_num(args[1])
if self.min_freq > self.max_freq:
self.min_freq, self.max_freq = self.max_freq, self.min_freq
```

swap them

```
# FIXME We set MANUALLY the physical limits of the device. In this
```

case

the USRP Flex2400 limits.

```
if self.min_freq < 2222000000:
print ("The minimum frequency of this device is 2.222GHz")
self.min_freq = 2222000000
if self.max_freq < 2222000000:
print ("The minimum frequency of this device is 2.222GHz")
self.max_freq = 2222000000
if self.min_freq > 2937000000:
print ("The maximun frequency of this device is 2.937GHz")
self.min_freq = 2937000000
if self.max_freq > 2937000000:
print ("The maximun frequency of this device is 2.937GHz")
self.max_freq = 2937000000
if self.min_freq == self.max_freq:
print ("Do not use this program for a single frecuency analysis
```

please")

exit()

```
self.fft_size = options.fft_size
if not options.real_time:
realtime = False
else:
# Attempt to enable realtime scheduling
r = gr.enable_realtime_scheduling()
if r == gr.RT_OK:
realtime = True
else:
realtime = False
print "Note: failed to enable realtime scheduling"
# If the user hasn't set the fusb_* parameters on the command
```

line,

# pick some values that will reduce latency.

```
if 1:
if options.fusb_block_size == 0 and options.fusb_nblocks ==
```

0:

if realtime: # be more aggressive

options.fusb_block_size =

gr.prefs().get_long(‘fusb’,

‘rt_block_size’, 1024)

options.fusb_nblocks =

gr.prefs().get_long(‘fusb’,

‘rt_nblocks’, 16)

else:

options.fusb_block_size =

gr.prefs().get_long(‘fusb’,

‘block_size’, 4096)

options.fusb_nblocks =

gr.prefs().get_long(‘fusb’,

‘nblocks’, 16)

```
#print "fusb_block_size =", options.fusb_block_size
#print "fusb_nblocks =", options.fusb_nblocks
# build graph
self.u = usrp.source_c(fusb_block_size=options.fusb_block_size,
fusb_nblocks=options.fusb_nblocks)
adc_rate = self.u.adc_rate() # 64 MS/s
usrp_decim = options.decim
self.u.set_decim_rate(usrp_decim)
usrp_rate = adc_rate / usrp_decim
self.u.set_mux(usrp.determine_rx_mux_value(self.u,
```

options.rx_subdev_spec))

self.subdev = usrp.selected_subdev(self.u,

options.rx_subdev_spec)

print “Using RX d’board %s” % (self.subdev.side_and_name(),)

```
s2v = gr.stream_to_vector(gr.sizeof_gr_complex, self.fft_size)
mywindow = window.blackmanharris(self.fft_size)
fft = gr.fft_vcc(self.fft_size, True, mywindow)
power = 0
for tap in mywindow:
power += tap*tap
c2mag = gr.complex_to_mag_squared(self.fft_size)
# FIXME the log10 primitive is dog slow
log = gr.nlog10_ff(10, self.fft_size,
```

-20*math.log10(self.fft_size)-10*math.log10(power/self.fft_size))

```
# Set the freq_step to 75% of the actual data throughput.
# This allows us to discard the bins on both ends of the
```

spectrum.

```
self.freq_step = 0.75 * usrp_rate
self.min_center_freq = self.min_freq + self.freq_step/2
nsteps = math.ceil((self.max_freq - self.min_freq) /
```

self.freq_step)

self.max_center_freq = self.min_center_freq + (nsteps *

self.freq_step)

```
self.next_freq = self.min_center_freq
# We define the minimum, maximum and frequency step in a global
```

statement to use them later.

```
global min_center_freq, max_center_freq, freq_step
min_center_freq = self.min_center_freq
max_center_freq = self.max_center_freq
freq_step = self.freq_step
tune_delay = max(0, int(round(options.tune_delay * usrp_rate /
```

self.fft_size))) # in fft_frames

dwell_delay = max(1, int(round(options.dwell_delay * usrp_rate /

self.fft_size))) # in fft_frames

```
self.msgq = gr.msg_queue(16)
self._tune_callback = tune(self) # hang on to this to
```

keep it

from being GC’d

stats = gr.bin_statistics_f(self.fft_size, self.msgq,

self._tune_callback, tune_delay,

dwell_delay)

```
# FIXME leave out the log10 until we speed it up
self.connect(self.u, s2v, fft, c2mag, log, stats)
#self.connect(self.u, s2v, fft, c2mag, stats)
if options.gain is None:
# if no gain was specified, use the mid-point in dB
g = self.subdev.gain_range()
options.gain = float(g[0]+g[1])/2
self.set_gain(options.gain)
print "gain =", options.gain
def set_next_freq(self):
target_freq = self.next_freq
self.next_freq = self.next_freq + self.freq_step
if self.next_freq >= self.max_center_freq:
self.next_freq = self.min_center_freq
if not self.set_freq(target_freq):
print "Failed to set frequency to", target_freq
return target_freq
def set_freq(self, target_freq):
"""
Set the center frequency we're interested in.
@param target_freq: frequency in Hz
@rypte: bool
Tuning is a two step process. First we ask the front-end to
tune as close to the desired frequency as it can. Then we use
the result of that operation and our target_frequency to
determine the value for the digital down converter.
"""
return self.u.tune(0, self.subdev, target_freq)
def set_gain(self, gain):
self.subdev.set_gain(gain)
```

def mean(data): # Returns the arithmetic mean of a

numeric

list

return sum(data) / len(data)

def main_loop(tb):

```
# We give basic information about the Spectrum Analysis
print "The start frequency is %s Hz" % min_center_freq
print "The final frequency is %s Hz" % max_center_freq
print "The frequency step is %s Hz" % freq_step
g = Gnuplot.Gnuplot(debug=1)
while 1:
# Get the next message sent from the C++ code (blocking call).
# It contains the center frequency and the mag squared of the
```

fft

m = parse_msg(tb.msgq.delete_head())

```
# Print center freq so we know that something is happening...
#print (m.center_freq)
# FIXME do something useful with the data...
# Mechanism to save in a file (power.dat) 2 columns, one for the
```

frequencies and the other for the mean of the FFT_SIZE points of m.data

```
if m.center_freq == min_center_freq: # If we get the minimum
```

frequency, it’ll reset the power.dat file

power=open(“power.dat”, “w”) # It will overwrite the

power.dat

file

```
power=open("power.dat", "a") # Each loop, it adds a dataline
```

(append)

p=str(m.center_freq) # with a frequency and the mean of

the

256 FFT samples (Power in dB)

media=str(mean(m.data)) #

todo= p + " " + media + ‘\n’ #

power.write(todo) #

```
if m.center_freq == (max_center_freq-freq_step): # If it gets the
```

final frecuency

```
p=str(m.center_freq) # It'll write the last
```

frecuency

with its Power in the power.dat file

media=str(mean(m.data)) #

todo= p + " " + media + ‘\n’ #

power.write(todo) #

g.load(“plot.p”) # Load the plot with the data

obtained from URSP

power=open(“power.dat”, “a”) # Without this line, the

file will start with the last frecuency

#g.hardcopy(‘spectrum.ps’, enhanced=1, color=1) # It does

a

plot copy to the hard disk (I think there’s not enough time to do it)

```
# m.data in 'w' mode: only write, if it exist a file with the same
```

name,

it’ll be overwrite.

# ‘a’ to append

# ‘r+’ for read and write

```
# m.data are the mag_squared of the fft output (they are in the
# standard order. I.e., bin 0 == DC.)
# You'll probably want to do the equivalent of "fftshift" on
```

them

```
# m.raw_data is a string that contains the binary floats.
# You could write this as binary to a file.
```

if **name** == ‘**main**’:

tb = my_top_block()

try:

tb.start() # start executing flow graph in another

thread…

main_loop(tb)

```
except KeyboardInterrupt:
pass
```

PLOT.P

##
set autoscale

unset logscale

unset label

set xtic auto

set ytic auto

set title “Wideband Spectrum Analyzer”

set xlabel “Frecuency”

set ylabel “Power (dB)”

set grid

plot “power.dat” using 1:2 title ‘Mean power’ with linespoints

View this message in context:

http://www.nabble.com/Wideband-Spectrum-Analyzer-tp19974701p20343196.html

Sent from the GnuRadio mailing list archive at Nabble.com.

See how the logpwrfft in gnuradio adjusts for different windows and fft

lengths:

http://gnuradio.org/redmine/repositories/entry/gnuradio/gnuradio-core/src/python/gnuradio/blks2impl/logpwrfft.py

-Josh